User guide • Read online or download PDF • KYORITSU A User Manual • KYORITSU Tools. Get KYORITSU A User guide. Get all KYORITSU manuals!. Find great deals for Kyoritsu Digital Loop Tester Model A. Shop with confidence on eBay!.
|Published (Last):||12 June 2005|
|PDF File Size:||8.10 Mb|
|ePub File Size:||14.19 Mb|
|Price:||Free* [*Free Regsitration Required]|
What’s missing? Tell us about it.
Every circuit must be tested to make sure that the fault loop impedance does not exceed that specified for the over current protection device concerned. Wiring check LEDs All material on this site is subject to copyright. Tell us what’s missing. Delivery times may vary, especially during peak periods. You have no items in your shopping cart. Tell us about it. The loop impedance meter also has a visual indication of reversed phase and neutral wiring t socket.
Don’t see a manual you are looking for? Be the first to review this product Ask Question. If you find inaccurate product information, welcome to Correct Errors. Such metalwork may become live for the duration of the test.
MODEL A｜LOOP/PFC/PSC Testers｜PRODUCTS｜KYORITSU
Shipping and handing Item location: Kyoritwu of the significant features of this model is the D-Lok circuit that permits loop impedance testing without tripping some types of RCD installed in the circuit being loop tested. Please allow additional time if international delivery is subject to customs processing.
Safe voltage measurement function. This advantage over many other loop impedance meters means there is no need to by-pass many RCD tyoes at distribution boards, when conducting Loop or PSC tests.
The A also allows direct reading kyorittsu the PSC prospective short circuit current. This mode is cancelled whenever the test button is pressed. If you are not quite sure how to proceed, stop and take advice from a qualified kyiritsu. Or point us to the URL where the manual is located. Page 13 In the event of a fault, the Fault loop impedance should be low enough and the Prospective Fault current higher enough in order to have the automatic disconnection of supply by the installed protection device within prescribed time interval.
It is fundamental for this example to test also the RCD to ensure that operation takes place quickly enough to respect the safety requirements. About StoreInfinity We source consumer electronics, electrical engineering tools and other electronic products. It is able to measure low loop resistance to kyoditsu resolution of 0.
This instruction manual contains warnings and safety rules which must be observed by the user to ensure safe operation of the instrument and retain it in safe condition.
Visual indication of reversed phase and neutral wiring at socket. Tags Add Your Tags: Write Your Own Review You’re reviewing: Contact Test and Tag Supplies. In the event of a fault, the Fault loop impedance should be low enough and the Prospective Fault current higher enough in order to have the automatic disconnection of supply by the installed protection device within prescribed time interval. Featured Category Find more Fluke products,You can use side bar filter to look for your need easier and fast.
Large custom digital display readout on all models. Rating A 6 10 16 20 25 32 40 50 63 80 Disconnecting time 5s Disconnecting time 0.
This convenient test instrument is capable of measuring fault loop impedance within the wiring circuit directly at the power outlet. If not, disconnect from the mains and check the wiring at the socket.
The current will not cause tripping out involved RCD even the one with the lowest nominal differential current 30mA. Analogue Multimeters Digital Multimeters Accessories. The RCD could trip during the tests also with model A.
The figure below shows in marked line the Fault loop impedance for TN system. Custom microprocessor controlled for highest accuracy and reliability. Via our supply 44120a management and product delivery to the customer’s local repository, we ensure our customers get affordable prices, fast kyoritsi and a logistics localized service Your manual failed to 41120a The cover can be detached and put on the back side of the main body during measurement.