We reserve rights to make modifications to this manual without any further notice. .. Use only cables provided by Beamex when connecting MC5 to a PC or a. BEAMEX MC5-IS | FIELD EQUIPMENT FIELD EQUIPMENT | BEAMEX MC5-IS .. CD-ROM with user manual, software tools and product information. MC5 is the first of its kind modular ALL-IN-ONE multifunction calibrator. Alternately, external and manual reference junction . Internet http://www.
|Published (Last):||6 March 2014|
|PDF File Size:||6.33 Mb|
|ePub File Size:||7.78 Mb|
|Price:||Free* [*Free Regsitration Required]|
The above not withstanding, we can assume no responsibility for any errors in this manual or their eventual consequences. We reserve rights to make modifications to this manual without any further notice. Other trademarks are property of their respective owners. There are however no definite restricitions; fill in the form when you feel like it all items need not be answered.
Then send it to Beamex using one of the possibilities listed below. Did anything in the product disappoint you? In that case, please specify. General Things discussed in Part A: MC5- IS that previously only included pressure modules. Thanks to the logical user interface MC5-IS is very easy to bdamex. The large graphical display guides the user in different languages and it displays results both numerically and graphically. It should be free of scrapes and scratches and in proper operation order upon receipt.
The receiver should, however, inspect the unit for any damage that may have occurred during transit. All sections and connections are presented in detail on geamex next pages. Keep in mind that the previous picture as well as all pictures of MC5-IS in this manual has an example configuration of modules. The configuration of your MC5-IS may vary significantly from the one in the picture.
PX1 in a sticker on the upper panel. One of them may be an internal barometric module.
The connectors for Internal Pressure modules start from the sec- ond connector on the left. The RS connector may be used when connecting to a serial port bexmex a PC. Part B of this manual describes in detail what kind of reference junction settings are available. Display MC5-IS has a reflective display.
BEAMEX MC5-IS USER MANUAL Pdf Download.
The resolution of the display is x pixels. To quickly tune the contrast of the display: All free memory may be used for anything that requires more memory.
Thus there is no exact limit for the number of instruments that MC5-IS can maintain in its memory. It all depends on how much memory is allocated by other data. A good average operating time is 5 hours. The whiter the picture is, the more acute is the need for re- Empty batteries: The charging electronics informs you of the phases of the charging procedure with the help of the charge status light. The Battery Pack has holes in order to vent generated gas and heat. Avoid exposing the Battery Pack to liquids.
It is mechanically prohibited because only an intrinsically safe Bat Therefore it is rela- tively easy to update the firmware whenever a new version with fresh capabilities is released.
Therefore very spe- cial attention was directed on this matter when creating the calibra- tor. MC5-IS may be used as a stand-alone calibrator i. MC5- The fourth section rightmost section displays additional informa- tion in the form of symbols, like: The meaning of the Function Keys varies depending on the situation. Menus The Function Key for opening 1 0. See also Part B for special features concerning numeric fields when generating a signal. You cannot add more digits if the length of the number is at its Delete maximum limit.
Use the Function Key to remove un- wanted digits first and then enter the new digits. You have to select one of the available options. MC5-IS includes several optional modules both in hardware and firmware. This makes it possible to buy a calibrator with capabilities according to current requirements. If additional needs arise later on, add more modules to your MC5-IS and you will have a tool that suits all demands.
So be careful when working in wet conditions. Use MC5-IS only if you are certain of that it can be used safely. Safe use of MC5-IS is no longer possible if one or more of the following cases are true: If you, however, connect together voltages from the ET and E sections or if you connect external voltages to MC5-IS, the re- sulting voltage may be high enough to be hazardous.
Only personnel with good experience and knowledge of high pressure liquid, air and ni- trogen operations are allowed to work with the module. Read carefully all these instructions and local safety instructions for high pressure operations before starting the use.
BEAMEX MC5-IS User Manual
There are, however a few things that anyone using MC5-IS may do. Wait a few minutes and then rinse using a cloth moistened with pure water.
Never use any strong detergents. After mv5 self test, some basic information of the calibrator at hand appears in the lower part of the screen. If you want to view the calibrator information for All non-calibration related measurements and generations are per- formed in the Basic Mode.
What can be done in Basic Mode If the module is not included in your MC5-IS, the Startup and Basic Operation Warning! IS is changed in the vertical direction. Both of the above men- tioned factors affect notably on the neamex measurement modules. Measuring pressure below mbar approx. Information con- cerning current measurement parallel to a test diode can be found in part C, chapter Current Measurement Parallel to a Test Diode, Connections on page Low voltage measurement can be used for non-standard thermo- couple measurement.
You will see the measured temperature in millivolts and need a table to convert the measured millivolt value to corresponding temperature values. R meas The following picture displays the active The two leftmost terminals are used in 2- terminals: Sound Also check the setting.
The option MC5-IS shows voltages above approx. A limit switch test displays the opening and closing point of a limit switch.
Unit Also check the setting. Edge Also check the setting. Available RTD sensors The following picture displays the active M5 two leftmost terminals are used in 2- terminals: Available only if is not used as the Reference Ebamex compensation method. TCx mea ule is an optional module. Startup and Basic Operation Note. MC5-IS has a comprehensive set of predefined sensor types. Addi- tional sensor types are available as options.
Special Measurements Special Measurements Mahual measurements are extra utilities that can be activated to perform a special function alongside the normal measurement.
Activating another special measure- ment deactivates the earlier special measurement in the same win- dow or the earlier special measurement using the same quantity. R a t e o f C h a n g e U n i t Special Measurements Rate of Change… Before the special measurement is started MC5-IS prompts for the rate of change unit which can be se- lected from the following options: In redundant measurement the measurements of the main port and the second port are compared with each other.
If the readings differ more than the entered limit value, MC5-IS gives an audible alarm.
Start the redundant measurement by selecting: If the module is not included in your This is done to prevent damages in the connected instrument. If an instrument that cannot withstand the generated signal is connected to MC5-IS, the instrument may get seriously dam- aged. MC5-IS acts as a device controlling the flow of current while a external device generates the supply voltage for the loop, both in the and the module.
If you open the mA generation loop, MC5-IS tries to maintain the current by decreasing the internal impedance. If you then close the loop again, the current is first too high, but returns quickly to the correct level.
Low voltage generation can be used when calibrating instruments with millivolt input. You can use it also to simulate special thermo- couples not defined in MC5-IS. If you simulate a thermocouple in millivolt mode, you must convert from temperature to millivolts manu- ally. Polarity Also check the setting. The instrument under test generates the current for the resistance measurement.
MC5-IS controls the voltage across nanual terminals so that the resistance voltage to current ratio corresponds to the simulated temperature or resistance. This property is called thermocouple simulation.
It enables checking and calibration of temperature indicators, recorders, transmitters and other equipment related to temperature measurement with a thermocouple probe.